FloWav PSA-AV Completes Testing at Alden Labs
Testing-Alden Labs: FloWav’s flagship sensor, the PSA-AV or Progressive Spectral Analysis-Area Velocity Sensor was tested by Alden Research Laboratory, Inc. The testing was completed in Alden’s hydraulics testing facility in Worcester, Massachusetts.
Alden’s standard test procedure, QMSM-01 Revision 4, was used for all aspects of the testing.
The sensor under test was installed in Test Line 3 in the Hooper Low Reynolds Number Facility. The Hooper Facility water is provided through a 40″ penstock from the main laboratory pond resulting in a gross gravity head of approximately 28 feet at a maximum flow of 35,000 gpm.
A calibrated Venturi was used to measure flow for Test Line 3.
The testing went well and the results were extremely favorable for the PSA-AV. FloWav is proud to say the sensor tested to an accuracy of +/-2% of reading for the Doppler crystals velocity reading and +/-0.25% full scale, +/-1% of reading from 32 deg. F to 160 def F for the ceramic pressure sensor’s depth readings. Results you can count on within your systems!
Click here for full procedure: Alden Summary FloWav 101311
Having confidence in your equipment is crucial. Let FloWav provide you with the best Flow Monitoring Equipment in the World.
We are proud to announce that the FloWav Service-Support staff has relocated their efforts to a new office in Chattanooga, TN. The new space for the FloWav Service-Support Staff includes newly renovated office space as well as a large warehouse where we are shipping...
New Flow Monitoring Equipment From FloWav - 2017 You will be blown away with our new flow monitoring equipment offerings. Stop by our booth, 4719, at WEFTEC to see the latest in technology and innovation! We live by our Value Statement: We focus on the one thing we...