FloWav PSA-AV Completes Testing at Alden Labs
Testing-Alden Labs: FloWav’s flagship sensor, the PSA-AV or Progressive Spectral Analysis-Area Velocity Sensor was tested by Alden Research Laboratory, Inc. The testing was completed in Alden’s hydraulics testing facility in Worcester, Massachusetts.
Alden’s standard test procedure, QMSM-01 Revision 4, was used for all aspects of the testing.
The sensor under test was installed in Test Line 3 in the Hooper Low Reynolds Number Facility. The Hooper Facility water is provided through a 40″ penstock from the main laboratory pond resulting in a gross gravity head of approximately 28 feet at a maximum flow of 35,000 gpm.
A calibrated Venturi was used to measure flow for Test Line 3.
The testing went well and the results were extremely favorable for the PSA-AV. FloWav is proud to say the sensor tested to an accuracy of +/-2% of reading for the Doppler crystals velocity reading and +/-0.25% full scale, +/-1% of reading from 32 deg. F to 160 def F for the ceramic pressure sensor’s depth readings. Results you can count on within your systems!
Click here for full procedure: Alden Summary FloWav 101311
Having confidence in your equipment is crucial. Let FloWav provide you with the best Flow Monitoring Equipment in the World.
Now through the end of October is a great time to purchase FloWav's Phaser V1000! All units are currently 20% off as we make room for 2021 inventory!!!!! The Phaser Model V1000, accurate, non-contact Doppler velocity sensor introduces digital Doppler radar...
Another high-reliability deployment of a RipTide R1-001 4G Cat-M1 / NB-IoT Ultra-low power Data Logger on an irrigation dam for one of our Gold Mining customers. The RipTide data logger is monitoring dam level and pumped outflow. The site records data at 15-minute...